TM 11 6625-1668-12
used for an eight level code character (Start-Stop and
The vertical deflection of the displayed
waveform is calibrated in VOLTS (MA)/CM. The
to the analyzer is either in HIZ or VCAL BRIDGING
(milliamperes) when the input to the analyzer is
through the SERIES jack, and the INPUT switch is in
either 60N, 20N, 20P, 30P or ICAL SERIES position.
The VOLTS (MA)/CM switch will measure either 2, 5,
10, 20 or 60 volts or milliampere depending on the
input to the analyzer. The last position marked 200
(volts only) will measure volts only. The range of the
VARIABLE control is 20% (continuously variable) of
the fixed positions of the VOLTS (MA)/CM switch.
(4) The VERTICAL POS Control allows the trace
to move vertically on the CRT.
(5) The HORIZONTAL POSITION adjusts the
horizontal position of the trace.
(6) The HORIZONTAL GAIN control is used to
adjust the horizontal size of the displayed waveform.
Figure 1-5. Cable Assembly, Special Purpose,
Figure 1-6. Cable Assembly, Special Purpose Electrical,